The Energy Research Centre for the Netherlands (ECN) has completed a 12 month offshore measurement campaign with the ZephIR 300 system as part of a four year campaign within the Dutch project “Meteorological Research Wind at Sea” where the ZephIR has been compared against the 108 metre Met Mast Ijmuiden (MMIJM) at the ECN Wind Turbine test facility EWTW. During a 60 week continuous period, the ZephIR was measured at over 98 percent, with the only period of unavailability being attributed to a mast power supply failure from which the ZephIR was being powered.
The data from the tests has been delivered to the wind energy industry on behalf of the Netherlands Ministry of Economic Affairs, Agriculture and Innovation. ECN’s initial conclusion is that ZephIR can be considered as a valid stand-alone system for wind resource assessments, especially given the industry’s tendency towards higher hub heights.
ECN’s Wind Energy Coordinator of Experiments & Measurements, Henk Oostrum noted, “ZephIR is very capable of providing high quality, 10 minute average wind speed and direction data at all heights well above traditional tall mast heights. Correlation is excellent to our IEC compliant mast and site. ZephIR can therefore be considered, in our initial opinion, to be used in a stand-alone application for wind resource measurements.”
Measurements at the EWTW and MMIJM are in accordance with IEC 61400-12. All masts are equipped with Thies class 1 cup anemometers and wind vanes as well as WindSensor p2456a cup anemometers. All calibrations are performed in a MEASNET approved wind tunnel and are covered by ISO 17025 ILAC accreditation.
“This is an exciting development in the ZephIR program – we never imagined that we would actually be operating within a met mast structure,” said Ian Locker, MD at Zephir Ltd. “The work that ECN has done independently, and now publicly, verifies not only the excellent core performance of ZephIR compared to a mast but also the ability to be deployed in this unique way. In both onshore and offshore applications, ZephIR has been proven to operate in a stand-alone manner providing finance-grade wind measurements for the progression of wind farm development and on-going performance analysis and optimisation.”